Preloading based needle insertion with a concave probe to enhance targeting in breast tissue
© Kobayashi et al.; licensee Springer. 2014
Received: 26 December 2013
Accepted: 7 October 2014
Published: 20 October 2014
Target displacement cased by tissue deformation is a major technical challenge in ultrasound-guided needle insertion for breast tumor treatment or biopsy. We previously developed a preloading-based needle insertion method, which consists of a preloading phase and an insertion phase, and validated its placement accuracy. The present study furthered this research by focusing on a new concave preloading probe to enhance needle insertion targeting. With this probe, we evaluated the several probe size for different tumor diameters and locations to investigate the geometry effect on insertion accuracy. In this in vitro study, preloading-based needle insertion with concave probe of the same diameter as that of the tumor showed the highest placement accuracy for shallow tumors, while concave probe of a larger diameter than that of the tumor showed the highest placement accuracy for deep tumors. Overall, the concave preloading probe described in this study decreased needle placement error. Future research focused on its evaluation by structural analysis based on soft tissue modelling and deformation simulation.
Breast cancer accounts for more than 1 million of the estimated 10 million malignancies diagnosed worldwide each year . Breast imaging technologies for early diagnosis of disease are improving along with treatments that may improve the prognosis . There has been a recent trend toward minimally invasive procedures such as needle biopsy and radiofrequency ablation (RFA) therapy. In these procedures, ultrasound (US) imaging is widely used to guide insertion of a needle into tumor tissue because it has the advantages of real-time visualization and cost-effectiveness. During the procedure, a clinician holds an US probe with one hand and inserts the needle with the other. In the case of RFA therapy, the needle then emits radiofrequency energy from microwaves to heat and destroy the tumor lesion. Alternatively, the needle takes a sample of the tumor for biopsy. For breast tumors, these therapies are promising as they are minimally invasive, cause less scarring, have quick recovery rates, and decrease cost.
However, technical challenges exist in needle insertion for breast tumor treatment. In spite of a real-time visible image from an US probe, accurate placement of the needle is still difficult, especially with small, early breast tumor lesions that require placement accuracy of only a few millimeters. Because the breast is soft, it can easily deform when a needle pierces it and advances toward an inner tumor. The deformation causes displacement of the tumor lesion. Thus, accurate needle insertion requires solving the problems of tissue deformation and target displacement.
1.2 Related research
Current research and development of robots and navigation systems for minimally invasive surgery is focused on improving needle insertion and placement -. For example, Bassan et al. reported a novel five-degrees-of-freedom manipulator for percutaneous needle insertion . A puncture-assisting robot operating under US guidance for use during procedures involving the prostate gland has also been developed . External actuators have been used to manipulate the position of breast lesions ,. In both studies, multiple actuators were used to decrease targeting error. In another study, a robotic medical needle insertion procedure using tissue manipulation was developed to improve targeting accuracy .
A number of research groups have also conducted studies to solve the problem of organ deformation associated with needle insertion . For example, methods have been developed for mechanical modeling of breast deformations , and simulation of steerable needle insertion for prostate therapy -. Deformation of a planar tissue phantom during needle insertion has been investigated using a linear elastic material model ,. Dehghan et al. determined the optimal angle of needle insertion and position using linear and nonlinear models . Picinbono et al. described nonlinear and anisotropic elastic liver models for medical simulation . A physics-based model for medical palpation of deformable tissue has also been reported . Many research groups have used animal organs as models to investigate measurement details associated with needle insertion, puncture and friction force, and tissue stiffness (e.g., -).
We have been researching the method to enhance the targeting in breast tissue, named a preloading-based needle insertion method. The concept of the method is that before inserting the needle, the breast is pressed by the preloading probe to stabilize the tumor as the needle approaches. During the needle insertion phase, the needle direction does not change to avoid damaging internal tissues. Insertion is stopped when the tumor position and needle tip are closest. Thus, this method has a preloading phase and a needle insertion phase to locate the preloading probe and pierce the breast tissue. The preloading phase hardens the tissue between the target and the needle tip to stabilize the tumor and make it easier to pierce the tissue during the needle insertion phase.
We previously reported this concept and its evaluation by numerical simulations and experiments both in vitro and in vivo. Smaller error with preloading-based insertion than with normal insertion was obtained in all experiments -.
1.4 Objectives and scope of this article
This section shows our evaluation of the preloading method with concave probe on needle insertion accuracy by in vitro experiments using a hog breast and in vivo experiments using a goat breast.
2.1 Experimental setup of in vitro experiments
2.1.1 Needle insertion manipulator
The manipulator had four degrees of freedom for planar movements and components for positioning and integrating the apparatus. The positioning component had three serial joints with rotational degrees of freedom. The integrating component had two parts with separate drives: one for the preloading probe and the other for needle insertion. Preloading probes with different diameters could be alternately attached. Each preloading probe had a center hole for insertion of a needle of 2.0 mm diameter. Six-axis force/torque sensors (NANO 1.2/1, BL AUTOTEC) were attached to the root of the probe and needle to measure the pressing force and reaction force. We calculated needle tip and preloading probe tip positions from the angle of each joint of the manipulator, measured by encoders attached to the motors.
2.1.2 Breast with mock tumor
We used flattened hemispherical hog breast tissue, approximately 100 mm in diameter and 20 mm thick. A silicone cylinder, 10 mm in diameter and 20 mm long, served as a mock tumor embedded in the breast tissue.
2.1.3 Tumor position recognition
To operate the preloading probe and determine the needle placement accuracy, we tracked tumor location using a camera and an image-processing computer. We manually set the tumor lesion on the camera image, and the computer automatically tracked the lesion using the pattern-matching method.
2.1.4 Tumor locations
2.1.5 Insertion path
For cosmetic reasons, the superior edge of the areola is a preferred location for initial insertion. Consequently, the top of the breast was selected as the initial insertion point for the experiments. The direction of needle insertion was downward from the upper half of the breast to the center of the tumor.
2.1.6 Condition of normal insertion
The needle was advanced at a constant velocity of 4 mm/s toward the center of the initial position of the tumor lesion. Insertion was complete when the tumor position and needle tip were closest. We measured the force loaded on the needle, the tumor position, and the needle tip position to determine the placement error. Four trial insertions were performed on four hog breast samples at two different tumor depths (total of 32 trials).
2.1.7 Condition of preloading-based needle insertion
Preloading-based needle insertion was conducted with three preloading probe diameters (10, 15, and 20 mm) with a concave shape. They were alternately attached during the experiments to compare the geometry effect of the preloading probe (Figure 2c). During the preloading phase, the preloading probe pressed the hog breast toward the tumor lesion. The probe pressed the breast at 4 mm/s as the probe axis was adjusted to the center of the tumor lesion using the camera information of tumor position. We stopped the preloading when the reaction force reached 2.0 N. We previously compared insertion accuracy with different loading forces and suggested 2.0 N as the appropriate pressing force that would enhance robustness in insertion accuracy . After the preloading phase, we moved to the needle insertion phase. We inserted the needle at a constant velocity of 4 mm/s. During the needle insertion phase, the needle direction did not change. Insertion stopped when the tumor position and needle tip were closest. Again, we measured the force loaded on the needle, the tumor position, and the needle tip position to determine the placement error. For each of the three probes of different diameters, four trial insertions were performed on four hog breast samples at two different tumor depths (total of 96 trials). They were performed in random order to avoid skewing the results.
2.1.8 Needle placement error
2.2 Experimental setup of in vivo experiment
We also evaluated the accuracy of preloading-based needle insertion in vivo. A goat breast was used as the sample because its stiffness is similar to that of the human breast and because the size of goats allows for relatively easy maneuvering compared with heavy animals such as porcine. We compared preloading-based insertion with both concave probe and normal insertion.
All animals received humane care in accordance with “Guide for the Care and Use of Laboratory Animals” published by the National Institute of Health (NIH publication 85–23, revised 1985), “Guidelines for Proper Conduct of Animal Experiments” formulated by Science Council of Japan (2006), and the guidelines determined by the Institutional Animal Care and Use Committee of Tohoku University.
2.2.1 Needle insertion manipulator
2.2.2 Goat breast with mock tumor
2.2.3 Experimental setup
Figure 5 shows the experimental setup of in vivo experiments. First, the surgeon searched the target inferior mock cancer by manipulating the US probe attached to the needle insertion manipulator. Next, the surgeon fixed the probe with the needle insertion manipulator using the endoscope holding device at the target location to visualize the target inferior mock cancer on the US image. Finally, the surgeon used the Graphical User Interface (GUI) to order the needle insertion path from the computer.
2.2.4 Target position measurement
To determine needle placement accuracy, we measured the target location using US equipment and an image-processing computer.
2.2.5 Needle and preloading probe position
We calculated the tip positions of the needle and the palpation probe from the angle of each joint of the manipulator, as measured by encoders attached to the motors.
2.2.6 Needle insertion error
The definition of insertion error was identical to that for the in vitro experiments.
2.2.7 Condition of normal insertion
We detected the target lesion using the US image and fixed the manipulator to place the lesion on the US image screen. The needle direction was straight downward from near the areola at a constant velocity of 4 mm/s toward the center of the initial position of the target lesion. Insertion was complete when the target position and needle tip were closest. We measured the force loaded on the needle, target position, and needle tip position to determine the placement error. We performed three trial experiments.
2.2.8 Condition of preloading-based insertion
The preloading probe pressed near the areola of the goat breast. The probe moved at 4 mm/s during the preloading phase with its position directed at the center of the target lesion on the probe axis. The target lesion was measured from the US image using the pattern-matching method. The preloading phase was finished when the force loaded on the preloading probe reached 0.75 N. A needle was then inserted at a constant velocity of 4 mm/s. Insertion stopped when the target position and needle tip were closest. We measured the force loaded on the needle, target position, and needle tip position to determine the insertion error. We performed three trial experiments.
3.1 In vitro experiment
3.1.1 Experiment with shallow tumor location
Results of statistical tests for insertion errors among different probe diameter in shallow tumor experiment
3.1.2 Experiment with deep tumor location
Results of statistical tests for insertion errors among different probe diameter in deep tumor experiment
3.2 In vivo experiment
3.2.1 US image during insertion
Figure 10 shows an example of the US image during the experiment. The image allowed for successful location of the target position.
3.2.2 Comparison with each insertion
4.1 Preloading-based needle insertion with concave probe
Preloading-based needle insertion with all concave probe diameters showed better placement accuracy than did normal needle insertion. This can be said under conditions of both short and long distances between the tumor and the initial insertion point. Our data from the in vivo experiment also suggests that pressing by the concave probe reduces needle placement error. The major reason to enhance needle insertion accuravy is same as the condition with convex shape probe  and described as follows: Many related works have stated that the main puncture event is designated by a peak force applied to the needle that is preceded by a steady rise followed by a sharp decrease ,. That is, when a needle punctures the breast, it does not immediately cut the tissue; instead, it first pushes the tissue and then punctures it. During this phase, tumor displacement is increased. The preloading method enhanced the needle insertion accuracy because the tumor displacement during needle insertion is decreasing. Our previous report also suggested that breast tissue with high strain displays nonlinear characteristics and higher stiffness parameters . Pressing by the preloading probe results in high strain in the tissue surrounding the tumor. This spreads the tissue to not only stabilize it, but also to produce a rapid increase in stress near the needle. Puncture thus occurs with little tumor displacement. The concave probe is especially supposed to harden the tissue surrounding the tumor as discussed in this section later. The other potential advantage of concave shape stabilize the tumor at right angles to the direction of needle insertion. The futher investigation should be carried out about this issue while the advantage was partially supported by our previous result .
4.2 Geometry effect of preloading probe
4.3 Advantage of preloading-based needle insertion
preloading method has following advantages: (1) The success rate is high because tumor displacement is small. (2) Tissue damage is minimal because the insertion depth is smaller and the needle line is straight, not steered through the tissue. (3) The procedure is fast because the same device sequentially presses and inserts the needle. Furthermore, compared with the tissue manipulation method  and , the system has potential to be small and cost-effective because few actuators are needed. These advantages are same between the probe with convex and cocave shape. Meanwhile, we have advocated preloading efficiency with concave probe to minimize tumor displacement during needle insertion. The concave probe is expected to harden the tissue surrounding the tumor more effectively and stabilize it at right angles to the direction of needle insertion, resulting in effective reduction of tumor displacement. Further investigation is required to clearly discuss the comparison of convex and concave probe in preloading method.
4.4 Simulation analysis
Major limitation of this study is about structural analysis based on soft tissue modelling. As shown in our previous research , the simulation analysis may reveals the several situation and condition discussed above. We have not carried out simulation analysis because our simulation does not include the appropreciate collision detection algorism between the probe and breat tissue, which is one of the most difficult problem in soft tissue modelling and simulation. Future research will focus on clarification of the tendency for a geometry effect of the preloading probe on placement accuracy with respect to tumor size and location based on simulation analysis. Moreover, the probe size can be optimized with respect to tumor size and location via repeated simulation anlysis with several probe size. The revealed tendency would also have the potential to suggest the new approach that a probe diameter changes in real time.
4.5 Boundary condition
The simplified experimental condition of the breast (i.e.. a half-cylinder shape with a single tissue type and simplified boundary condition) is one of the limitations of the present work. We plan to develop a system with heterogeneous patient-specific modeling, which incorporates features such as shape, regions of each tissue and boundary conditions, using preoperative CT/MRI diagnostic images. Moreover, the collision between ultrasound probe and breast should be included in these boundary condition setteing.
We have described a needle insertion method involving tissue preloading to provide accurate needle insertion. The mechanical preloading probe locates tumor tissue and reduces tissue displacement during needle insertion by pressing the breast tissue. This paper focuses on investigation to evalauate preloading concept with concave probe to enhance accurate needle insertion. We compared preloading-based needle insertion using three concave probe diameters among one another and with normal insertion by experiments in vitro under conditions of two different tumor depths. Preloading-based needle insertion with a probe of the same diameter as that of the tumor showed the highest placement accuracy for the shallow tumor location. On the other hand, preloading-based insertion with a probe of larger diameter than that of the tumor showed higher placement accuracy for the deep tumor location. These results assumed that the appropriate preloading probe diameter could be determined with respect to tumor diameter and location. Our data from the in vivo experiment also suggest that pressing by the concave probe reduces needle placement error.
We will also research the integration of the palpation system into our system. A preloading probe will be used for both palpation and preloading. Reaction force applied to the preloading probe will indicate the tumor direction. Our goal is to construct a surgical robot that incorporates both the needle insertion and palpation systems for clinical application.
This work was supported in part by the Global Centers of Excellence Program “Global Robot Academia,” Waseda University, Tokyo, Japan; High-Tech Research Center Project from the Ministry of Education, Culture, Sports, Science and Technology; a Grant-in-Aid for Scientific Research (A) (23240081); and a Grant-in-Aid for Young Scientists (B) (23700585).
- Ferlay J, Bray F, Pisani P, Parkin DM (2001) GLOBOCAN 2000: cancer incidence, mortality and prevalence worldwide. IARC CancerBase :5Google Scholar
- Vlastosa G: Minimally invasive approaches for diagnosis and treatment of early-stage breast cancer. Oncologist 2007, 12: 1–10. 10.1634/theoncologist.12-1-1View ArticleGoogle Scholar
- Kazanzides P, Fichtinger G, Hager GD, Okamura AM, Whitcomb LL, Taylor RH: Surgical and interventional robotics. IEEE Robotics & Automation Magazine 2008, 15: 122–30. 10.1109/MRA.2008.926390View ArticleGoogle Scholar
- Fichtinger G, Kazanzides P, Okamura AM, Hager GD, Whitcomb LL, Taylor RH: Surgical and interventional robotics: part II. IEEE Robotics & Automation Magazine 2008, 15: 94–102. 10.1109/MRA.2008.927971View ArticleGoogle Scholar
- Hashizume M, Yasunaga T, Tanoue K, Ieiri S, Konishi K, Kishi K, Nakamoto H, Ikeda D, Sakuma I, Fujie M, Dohi T: New real-time MR image-guided surgical robotic system for minimally invasive precision surgery. Int J Comp Ass Rad Surg 2008, 2: 317–25. 10.1007/s11548-007-0146-9View ArticleGoogle Scholar
- Bassan HS, Patel RV, Moallem M: Novel manipulator for percutaneous needle insertion: design and experimentation. IEEE/ASME Trans Mechatron 2009, 14: 746–61. 10.1109/TMECH.2009.2011357View ArticleGoogle Scholar
- Fichtinger G, Fiene JP, Kennedy CW, Kronreif G, Iordachita I, Song DY, Burdette EC, Kazanzides P: Robotic assistance for ultrasound-guided prostate brachytherapy. Med Image Anal 2008, 12: 535–45. 10.1016/j.media.2008.06.002View ArticleGoogle Scholar
- Vishnu G, Sarkar N, Podder TK (2008) Robot assisted real-time manipulation for breast biopsy. IEEE Proc 2008 Int Conf Robot Automation :2515–2515Google Scholar
- Mallapragada VG, Sarkar N, Podder TK: Robot-assisted real-time tumor manipulation for breast biopsy. IEEE Trans Robot 2009, 25: 316–24. 10.1109/TRO.2008.2011418View ArticleGoogle Scholar
- Torabi M, Hauser K, Alterovitz R, Duindam V, Goldberg K (2009) Guiding medical needles using single-point tissue manipulation. Proc 2009 IEEE Int Conf Robot Automation :2705–10Google Scholar
- Larson BT, Erdman AG, Tsekos NV, Yacoub E, Tsekos PV, Koutlas IG: Design of an MRI-compatible robotic stereotactile device for minimally invasive interventions in the breast. J Biomech Eng 2004, 126: 458–65. 10.1115/1.1785803View ArticleGoogle Scholar
- Azar FS, Metaxas DN, Schnall MD: A deformable finite element model of the breast for predicting mechanical deformations under external perturbations. Acad Radiol 2001, 8: 965–75. 10.1016/S1076-6332(03)80640-2View ArticleGoogle Scholar
- Azar FS, Metaxas DN, Schnall MD: Methods for modeling and predicting mechanical deformations of the breast under external perturbations. Med Image Anal 2002, 6: 1–27. 10.1016/S1361-8415(01)00053-6View ArticleGoogle Scholar
- Alterovitz R, Goldberg K, Okamura A (2005) Planning for steerable bevel-tip needle insertion through 2D soft tissue with obstacles. Proc 2005 IEEE Int Conf Robot Automation :1652–7Google Scholar
- Alterovitz R, Goldberg KY, Pouliot J, Hsu IC: Sensorless motion planning for medical needle insertion in deformable tissues. IEEE Trans Inf Technol Biomed 2009, 13: 217–25. 10.1109/TITB.2008.2008393View ArticleGoogle Scholar
- Chentanez N, Alterovitz R, Ritchie D, Cho J, Hauser K, Goldberg K, Shewchuk JR, O’Brien JF: Interactive simulation of surgical needle insertion and steering. ACM Trans Graph 2009, 88: 1–10. 10.1145/1531326.1531394View ArticleGoogle Scholar
- DiMaio SP, Salcudean SE: Needle insertion modelling and simulation. IEEE Trans Robot Automation 2003, 19: 864–75. 10.1109/TRA.2003.817044View ArticleGoogle Scholar
- DiMaio SP, Salcudean SE: Interactive simulation of needle insertion model. IEEE Trans Biomed Eng 2005, 52: 1167–79. 10.1109/TBME.2005.847548View ArticleGoogle Scholar
- Dehghan, E, Salcudean, SE 2007Needle insertion point and orientation optimization in non-linear tissue with application to brachytherapy226772Google Scholar
- Picinbono G, Delingette H, Ayache N (2001) Nonlinear and anisotropic elastic soft tissue models for medical simulation. Proc 2001 IEEE Int Conf Robot Automation :1370–5Google Scholar
- Daniulaitis V, Alhalabi MO, Kawasaki H, Tanaka Y, Hori T (2004) Medical palpation of deformable tissue using physics-based model for haptic interface robot. Proc 2004 Int Conf Intell Robots Systems :3907–11Google Scholar
- Okamura AM, Simone C, O’Leary MD: Force modeling for needle insertion into soft tissue. IEEE Trans Biomed Eng 2004, 51: 1707–16. 10.1109/TBME.2004.831542View ArticleGoogle Scholar
- Heverly M, Dupont P, Triedman J (2005) Trajectory optimization for dynamic needle insertion. Proc 2005 IEEE Int Conf Robot Automation :1646–51Google Scholar
- Mahvash M, Dupont P (2009) Fast needle insertion to minimize tissue deformation and damage. Proc 2005 IEEE Int Conf Robot Automation :3097–102Google Scholar
- Kobayashi Y, Suzuki M, Konishi K, Hashizume M, Fujie MG (2008) Development of a novel approach, ‘palpation based needle insertion,’ for breast cancer treatment. Proc 2008 IEEE Int Conf Robotics Biomimetics :505–11Google Scholar
- Kobayashi Y, Suzuki M, Kato A, Konishi K, Hashizume M, Fujie MG (2009) A robotic palpation-based needle insertion method for diagnostic biopsy and treatment of breast cancer. Proc 2009 IEEE Int Conf Intell Robots Systems :5534–9Google Scholar
- Hatano M, Kobayashi Y, Suzuki M, Shiraishi Y, Yambe T, Konishi K, Hashizume M, Fujie MG (2011) In vitro and in vivo validation of robotic palpation-based needle insertion method for breast tumor treatment. Proc 2011 IEEE Int Conf Robot Automation :392–7Google Scholar
- Kobayashi Y, Suzuki M, Kato A, Hatano M, Konishi K, Hashizume M, Fujie MG: Enhanced targeting in breast tissue using a robotic tissue preloading-based needle insertion system. IEEE Trans Robot 2012,28(3):710–722. 10.1109/TRO.2012.2183055View ArticleGoogle Scholar
- Kobayashi Y, Hong J, Hamano R, Okada K, Fujie MG, Hashizume M: Development of a needle insertion manipulator for support during central venous catheterization. Int J Med Robot Comp Assist Surg 2011,8(1):34–44. 10.1002/rcs.420View ArticleGoogle Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited.